Cleaning and Storage - Air EEG
Proper procedure for cleaning, maintaining and storing your Air EEG system
To clean the EEG headset, use a cloth dampened with water.
Note: Do not use aggressive or abrasive cleaners or shampoo to clean the headset. Never submerge the system in liquid.
To disinfect the Air system, we recommend using Clinell Universal Wipes. Wipe down surfaces so that they remain damp for approximately one minute. Do not use excess pressure so that too much moisture is released at once.
In the area of the sensors, wipe with the sensors pointed downward so that excess moisture does not seep into the device.
Read disinfectant solution instructions carefully and confirm compatibility with the headset material. Bitbrain does not recommend the use of steam-based sterilization methods.
Keep the Air system on its provided stand or in its original packaging and in a dry place when not in use.
If the device will not be used for an extended period, we recommend discharging the battery to 40-50% capacity. Storing the device for more than a week at 100% charge could cause permanent capacity loss.
Recommended battery storage temperature range:
-10ºC to 35ºC (14ºF to 95ºF) for up to 1 month
-10ºC to 30ºC (14ºF to 86ºF) for up to 6 months